The ForMAX beamline will be equipped with two horizontally deflecting monochromators: either a “high-resolution” double crystal monochromator (hDCM) or a “high-flux” double multilayer monochromator (hDMM). The beam will pass onto dynamically bendable mirrors (VFM and HFM) in Kirkpatrick-Baez geometry, providing harmonic rejection and the possibility for independent focusing/collimation in vertical and horizontal directions. The optics hutch will also include four diagnostics units.
Secondary optics based on compund refractive lenses (CRLs) will be installed in the experimental station, providing the possibility of further increasing or decreasing the beam size.
|X-ray source||In-vacuum undulator (IVU), 3 m magnetic length, 17 mm period, 166 periods, 4 mm minimal magnetic gap.|
|Energy range||8-25 keV, using 3rd-13th harmonics of the IVU.|
|Crystal monochromator (hDCM)||Cryo-cooled horizontally deflecting double-bounce Si(111) crystal monochromator with fixed exit. Positioned 27 m downstream of the source.|
|Multilayer monochromator (hDMM)||Water-cooled horizontally deflecting double-bounce multilayer monochromator with fixed exit. W/B4C and Ru/B4C coatings. Positioned at 25 m.|
|Focusing mirrors||Vertically (VFM; 30.2 m from source) and horizontally (HFM;31 m from source) bouncing mirrors in Kirkpatrick-Baez geometry. Dynamically bendable flat mirrors, allowing focusing the beam at various planes down to sample position, collimating the beam, or no focusing, independently in vertical and horizontal directions (bending radii 5-100 km). Si, Rh, and Pt stripes for harmonic rejection.|
|Sample position||42 m downstream of the source.|
|Flux at sample position||1013 - 10 14 ph/s using the hDCM, 1014 - 10 15 ph/s using the hDMM.|
|Beam size at sample position (FWHM; vertical x horizontal)||Variable between 1x1.5 mm2 and 20x40 µm 2. Beam sizes up to 5x5 mm2 or down to 1x2 µm 2 possible using compound refractive lenses, to be installed in the experimental station.|